Regimen Upkeep Of The Glove Box
Etelux reminds you that in addition to the normal inspection of the glove box, the daily upkeep operate of the glovebox is also important
1. Just before and after every single do the job, 1st cleanse the surroundings to make the floor dry and cleanse. Then cleanse the tools, including the interior and outer surfaces of the glove box, the observation process, the do the job surface area, and many others., to be totally free of particles and thoroughly clean.
2. Very carefully record the drinking water, oxygen material, and fuel intake of the tools in the glove box just about every day to keep away from inadequate gasoline force and induce the products to quit operating.
three. Fork out attention to the quantity of gas in the doing work gasoline cylinder at any time and substitute it in time.
four. In situation of power failure, press the "vacuum pump" button on the panel to make the vacuum pump run after ability on
5. Note: After electric power on, if there is a dilemma that it cannot start off, the chance is that the vacuum pump energy cable is loosely connected. The 2nd purpose is that the vacuum pump’s personal change is not turned on
6. When you want to place the sample into the glove box, use a smaller changeover chamber as considerably as probable. Prior to opening the changeover doorway, pay back interest to regardless of whether the pressure gauge pointer on the valve cavity is at the zero position. If it is, you can open up it directly if not, you need to turn the change underneath the valve cavity to the "filling" placement to permit the gasoline enter the glove box transition compartment, and then switch the change again to the intermediate position, then you can open up the changeover compartment Put the sample in the door, and then shut the doorway. When inserting samples, be mindful not to touch the interior wall of the changeover chamber with your arms, use gloves to decide on and place the sample
Carrying out these regime servicing responsibilities can lower the decline level of the glove box, increase the services daily life of the glove box, and also minimize the prevalence of mishaps